OPHTHALMIC SURGERY CHAIRMEPRO IV SERIES
The ophthalmic surgery chair MEPROIV has the dominant market share in Japan. The space-saving chair for surgery relieves patient’s feeling of uneasiness and offers usability for surgeon and surgical staff.
The finely adjustable backrest and headrest hold respective patient’s figure and head shape.
Prompt shift to the surgical position
The chair tilts to the surgical position and returns to the position for the patient’s leaving smoothly and quietly by stepping on a switch on the foot control.
Angle adjustment of the headrest
Natural movement of the headrest according to the movement of the patient’s neck is realized by interlocking tilt and contraction/extension of the headrest.
Backrest height adjustment
Wide adjustment range of backrest being compatible with both small patients and tall patients
Armrests can be moved down for easy transfer of patients
from a wheelchair or stretcher/
Headrest cushion lifting
Height can be adjusted to two levels so that
it is compatible with slightly hunchbacked patients.
Surgical chair for cutting from the temple side MEPRO 4J
Provides a wider working area around the doctor’s feet so that operation of the foot control is not restricted in the surgery from temple side.
Dotted line: Backrest of MEPRO IV
In order to reduce interference with mobility of the surgeon’s knees, the backrest is thinner than the same of MEPRO IV.
|MEPRO IV||MEPRO 4J|
|Lifting and lowering the chair||465-730mm (Electro-hydraulic system by foot control)||485-770mm (Electro-hydraulic system by foot control)|
|Erecting and reclining the backrest||0-80° (Electro-hydraulic system by foot control)||0-80° (Electro-hydraulic system by foot control)|
|Backrest height||100mm by manual operation||–|
|Extending and retracting the headrest||150mm (Electro-hydraulic system by foot control)||170mm (Electro-hydraulic system by foot control)|
|Bending forward and backward the headrest||35°(Electro-hydraulic system by foot control)||35°(Electro-hydraulic system by foot control)|
|Locking and unlocking the chair||Electro-hydraulic system operated by the Chair Lock/Unlock switches||Electro-hydraulic system operated by the Chair Lock/Unlock switches|
|Product weight||150 kg||155 kg|